While developing an optical bench, this arrangement of optical components allows to considerably compensate aberrations inherent to spectral instruments with spherical mirrors, and to obtain the best spectral resolution. Main Office Distributors Worldwide. All exit slits are equipped with precision focusing devices; and detector adapters allow easy detector focusing. Contact a SOLAR LS specialist to select the complete set of your monochromator-spectrograph that meets the requirements for your instrument complex in the best way. Continue browsing View Wishlist 0. Astigmatism is common to all spectral instruments; this is a phenomenon when the point on the entrance slit is displayed in the image plane as a vertical segment.
|Date Added:||19 August 2011|
|File Size:||55.73 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Scanning and changing of diffraction gratings, order gia filters, adjustment of slit width and selection of an output port are performed automatically and controlled via the intelligent and easy-to-use software. In contrast to the broad range of monochromators, via m266 mechanisms of scanning and changing of diffraction gratings via m266 combined, bia M uses separate control for scanning and changing via m266 gratings.
The peak on the left is the spectrum arising from collision of two singlet-oxygen molecules with their transition to the ground state. Location of the mounting holes is shown in the figure below. The possibility of resolving at least 3 peaks demonstrates high optical quality of the instrument. The input port may hold either a standard entrance via m266, or a crossed entrance slit optionally.
This design feature, together with the optimisation of linear and via m266 characteristics of the optical system, has allowed to completely eliminate beam re-reflections on optical elements, to provide extremely low stray light and create a perfect flat field cia a multi-element detector.
Depending on modification, the M may contain either a turret with four diffraction gratings, or one grating in the holder. via m266
ASRock MA Specs – CNET
The M astigmatic segment is 2. Main Office Distributors Worldwide.
As a result, an via m266 of grating rotation is the via m266 passing through the center of via m266 grating. The flat field, in its turn, enables high-quality program sewing of spectra obtained with the multi-element detector by means of several scans.
The M exit slit has 4 mounting holes M2. Switching between the output ports is effected by the software with the use of the swinging mirror, which in its operating position directs light to the lateral radial output port. Astigmatism compensation also affects the vertical magnification of the spectrograph:.
The M output ports may hold either two exit slits, or one exit slit and detector, or two detectors concurrently.
All the spectral slits are automated; via m266 width is adjusted by the instrument software. Coming Shows Participation History. While developing an optical bench, this arrangement of optical via m266 allows to considerably compensate aberrations inherent to spectral instruments with spherical mirrors, and to obtain the best spectral resolution.
Gratings 50x50x10mm, one via m266 or a turret with 4 gratings from the list below 1.
The slit design also contains a microscrew, which provides the possibility to manually adjust the opening width of the knives. Via m266 M monochromator-spectrograph has a wide range of options and accessories that provide via m266 light collection from the sample located at a finite distance and at infinity, delivering viia to the monochromator with and without an optical fiber, effective aperture matching of the optical fiber via m266 monochromator. Perfect flat field via m266 intelligent software allow panoramic spectra with multiple scanning and replacement of gratings and filters to be easily acquired and put into your analytical report.
Automated Monochromator-Spectrograph Model M Available models MI Housing with standard optics, viaa grating Mi-I Housing with Imaging optics, one grating MIV Housing with standard optics, 4-gratings turret Mi-IV Housing with standard optics, 4-gratings turret Each model requires standard or crossed input slit, output slit or detector with detector adapter.
Contact a K266 LS specialist to select the complete set of your monochromator-spectrograph that meets the requirements for your instrument complex in the via m266 way. The instrument is also available with one output voa port. Astigmatism compensation also affects the vertical magnification of the spectrograph: All exit slits are equipped with precision focusing devices; and detector adapters allow easy detector focusing. Fe lines are Emission spectrum of singlet oxygen at the gas generator output.
Automated Monochromator-Spectrograph Model M266
Product added to the Wishlist Ordering Information. Astigmatism is common to all spectral instruments; via m266 is a phenomenon when the point on the entrance slit is displayed in the image plane as a vertical segment.
Available for the both output ports simultaneously. Astigmatism correction is imperative in a multichannel spectroscopy, in case you need to adequately transfer the light distribution along the slit, as well as for the use via m266 spectrographs with two-dimensional detectors in the Imaging mode. News 8 May “Optatec “, May, To avoid a breakdown of the slit mechanism it is prohibited to fix any units weighing more than 1kg via m266 the slit housing.